What is Dew Point?
The Dew Point is the lowest temperature that allows water vapour to remain in a gas without condensing to a liquid state. As the air or gas temperature drops, its ability to absorb water vapour drops until it becomes completely saturated and below this dew point temperature, water droplets will start to form.
In pressurised systems such as compressed air distribution networks, the dew point is directly related to the temperature AND system pressure. As pressure increases, the dew point temperature also rises meaning that the potential for vapour to condense out occurs at a higher temperature. In practice this can mean that instead of the condensing temperature being quite low, the dew point could be at or above ambient temperature.
Why do I need Dew Point Measurement?
Industrial compressed air and gas systems can suffer damage from water contamination directly or through the water subsequently freezing and expanding. Having water vapour in the air or gas can also affect process or product quality. Removing water contamination through filters and drying systems is common practice but as the dew point (and potential for damaging condensation) changes with pressure, the risk of damage will vary throughout a plant.
ISO 8573-1 defines a range of purity classes for compressed air including water with the measurement expressed in terms of pressure dew point. ISO 8573-3 is the section that defines measurement methods for humidity and ISO 8573-9 for liquid water.
Measuring the dew point can be a simple process and should be continuously monitored according to the plant risk analysis recommendations. This needs to occur at the drying plant before the distribution network and also at critical points of use. By measuring the dew point, efficient control of the drying/filtering system can be achieved to optimise the running costs of the compressed air/gas system.
Particular emphasis should be placed on any part of the system where the pressure increases or the ambient temperature decreases as these are the situations where problems can occur quickly. For example, the air ring main leaves one building to go to another and the ambient outside air is or can be significantly lower than the indoor ambient. Additional drying capability may be needed and monitored in order to prevent condensation as the pipe leaves the building.
Monitoring the dew point is a key indicator of system health that can be used to guide maintenance tasks as well as maintain plant compliance with the purity classes set in ISO 8573-1.
S201 Dew Point Sensor with Display
S212 -50degC Dew Point Sensor
S215 -20degC Dew Point Sensor
S217 OEM Dew Point Sensor
S220 -100degC Dew Point Sensor
S230 - S231 -100degC Dew Point Sensors
S305 Dew Point Monitor
S505 Handheld Dew Point Meter
ISO 8573 Air Purity
For a deeper understanding of which valve or instrument would be best for you please call or email us now so that we can save you time and ensure you can make a truly objective decision for your company.
Need assistance with a product or an enquiry? Fill in the form below and we will get in touch with you as soon as possible: